If it helps further, I highlighted the execution of the hpA. The reset might take it out of SYST: Obsolete View Service Options. OpenDocument ] I tried the hpa Initialise with Options. First, there is no soft front panel, as the HP A can be controlled from it’s hardware front panel. Using Labview control Agilent A signals generated occurs an error occurs. Some older versions of the IB Libraries provided this header file during installation.
|Date Added:||1 August 2005|
|File Size:||36.64 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The HP A driver is located in a self-extracting archive file called hpa. Message 2 of 7. Remove the fuse-holder assembly with the blown labvview from the rear, and install the new fuse, Then the Configure Hp a labview VI is called to set the instrument up for the desired measurement.
Is the code for hpA.
Message 5 of But one thing you can do is use Event Structure. Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type.
HP A LABVIEW DRIVER FOR WINDOWS 7
Thanks for your time Les. The only hp a labview are that it does not have a soft front panel or a knowledge base file. Introduction The help file included with the HP A driver contains instrument and programming information.
How do I change the line voltage setting on the A? For Windows NT, change references to win95 to winnt.
HP 33120A LABVIEW DRIVER
Using Labview control Agilent A signals generated occurs an error occurs. I’m guessing the changing of the waveform shape causes the dropout. Both examples first turn off the output before turning it back on with a different amplitude. But I get an laview message actually on the fg when the initialise vi is run in Labview. After the instrument has been configured, the Read VI is used to read multiple points.
OpenDocument ] I tried the hpa Initialise with Options. Message 4 of Accepted by topic author jto2.
HP 33120A LABVIEW DRIVER FOR WINDOWS 7
My driver used SYST: There are several advantages to using the dll form of the driver. Some older versions of the IB Libraries provided this header file during installation. All of references which follow assume Windows I’ve attached images from my scope that show what happens when I do this.
Wait another 10 mSec before sending another command or query. Second, there is no knowledge base file, which is primarily a physical description of a VXI board.
If you do find a command that is causing the instrument hp a labview report an error, contact NI so that they can fix the bug. Remove the fuse-holder assembly with the blown fuse from the rear, and install the new fuse, It would help if Hp a labview would try the Init function with the hp3120a options and see which is causing the error.
Message 4 of 7.
What is inside of the VIs that set the amplitude? Mean time between failure MTBF is 40, hours. Thinking back to when I used Agilent RF Generators, I could just set the amplitude or frequency and it wouldn’t turn off on me.
Do not include hpa. REM” after the reset. Now the waveform changes amplitude without turning off the output.